Product parameters:  Brand: FLUKE / Fluke
 Model: 411D
 Item: Industry
 COLOR CLASS: 411D 414D 419D 424D
 Decoration and construction content: installation works
 How it works: Laser
 Distance measuring distance: 30M
Buy Fluke any product to send a practical computer mouse pad, first come first served! Technical Parameters  Distance (for longer distances, use one target)  Fluke 411D  0.1 m to 30 m   (0.33 ft to 100 ft)  Fluke 416D  0.05 m to 60 m   (0.16 ft to 200 ft)  Fluke 421D  0.05 meters to 100 meters   (0.16 to 330 feet)   Measurement accuracy  Fluke 411D *  ± 3 mm (0.118 in)  Fluke 416D **  ± 1.5 mm (± 0.059 in)  Fluke 421D  ± 1.5 mm (± 0.059 in)   Laser level  Fluke 411D  Class II  Fluke 416D  Class II  Fluke 421D  Class II   Laser type  Fluke 411D  635 nm, <1 mW  Fluke 416D  635 nm, <1 mW  Fluke 421D  635 nm, <1 mW   Automatic power off  Fluke 411D  After 180 seconds  Fluke 416D  After 180 seconds  Fluke 421D  After 180 seconds   Continuous measurement  Fluke 411D  Yes  Fluke 416D  Yes  Fluke 421D  Yes   Plus / minus  Fluke 411D  Yes  Fluke 416D  Yes  Fluke 421D  Yes   Battery Life  Fluke 411D  Up to 3000 measurements  Fluke 416D  Up to 5000 measurements  Fluke 421D  Up to 5,000 measured values   LCD lighting  Fluke 411D  no  Fluke 416D  Yes  Fluke 421D  Yes   The data location  Fluke 411D  no  Fluke 416D  10  Fluke 421D  20   Minimum / maximum value  Fluke 411D  no  Fluke 416D  Yes  Fluke 421D  Yes   Sound feedback  Fluke 411D  no  Fluke 416D  Yes  Fluke 421D  Yes   Pythagorean function (indirect measurement)  Fluke 411D  simple  Fluke 416D  all  Fluke 421D  complete   Degree of protection  Fluke 411D  IP40  Fluke 416D  IP54  Fluke 421D  IP54   size  Fluke 411D  123 mm x 50 mm x 26 mm   (4.84 in x 1.97 in x 1.02 in)  Fluke 416D  135 mm x 46 mm x 31 mm   (5.31 in x 1.81 in x 1.22 in)  Fluke 421D  127 mm x 52 mm x 29 mm   (5.00 inches by 2.06 inches by 1.13 inches)   weight  Fluke 411D  150 g (5.29 oz)  Fluke 416D  110 g (3.88 oz)  Fluke 421D  125 grams (4.40 ounces)   temperature range  Fluke 411D  Storage: 25 ° C to 70 ° C (13 ° F to 158 ° F)   Operating: 0 ° C to 40 ° C (32 ° F to 104 ° F)  Fluke 416D  Storage: 25 ° C to 70 ° C (13 ° F to 158 ° F)   Operating: 0 ° C to 40 ° C (32 ° F to 104 ° F) 
Fluke 421D

25 ° C to 70 ° C (13 ° F to 158 ° F)
  0 ° C to 40 ° C (32 ° F to 104 ° F)   Use altitude (ISO 9022)
 Fluke 411D  Up to 3500 m  Fluke 416D  Up to 3500 m  Fluke 421D  Up to 3500 meters   Storage humidity (At 35 ° C)
 Fluke 411D  Up to 85% (24 hours)  Fluke 416D  Up to 85% (24 hours)  Fluke 421D  Maximum 85% for 24 hours   battery  Fluke 411D  AAA (2)  Fluke 416D  AAA (2)  Fluke 421D  AAA (2)   * Up to 10 m (33 ft) under favorable conditions (best target surface, room temperature) Under adverse conditions (eg strong sunlight, weakly reflecting target surfaces or large temperature fluctuations), distances of 10 m (33 ft) or more, the measurement accuracy may decrease to approximately ± 0.25 mm / m (± 0.003 in / ft). ** Up to 10 m (33 ft) under favorable conditions (best target surface, room temperature). Under adverse conditions (eg strong sunlight, weakly reflecting target surfaces or large temperature fluctuations) The deviation at m (33 ft) may be increased by approximately ± 0.15 mm / m (± 0.0018 in / ft).     The 414D reduces errors, saves time and reduces costs. characteristic:  Measuring the distance of the most advanced laser technology
 With onetouch operation, the distance between two objects can be measured instantly
 Measuring distances up to 50 m (165 ft) with an accuracy of 2 mm (0.08 in)
 Even at longer distances, bright lasers can be easily positioned
 Fast calculation of area (sq ft / m) and volume
 Simple addition and subtraction operation
 Using the other two measurements, the Pythagorean theorem can be used to measure the height indirectly
 Up to five measurement results can be stored
 Battery life for 3000 measurements, with autooff function
 Two lines, and has a continuous display (until the next button)
 The carrying bag is secured to the tool band
 Powered by two AAA batteries
 All of the above instruments have the following functions and advantages:  Reduce the estimation error, save time and cost
 Measuring the most advanced level 2 laser technology
 Just onetouch operation to complete realtime measurements
 Simple addition and subtraction operation
 Quickly calculate area (building area) and volume
 Addition and subtraction measurements are simple and convenient
 Min / Max function
 The autooff function increases battery life
 The distance from the other two measuring points is indirectly determined by the Pythagorean method
 Carrying bag with Fluke logo
 Threeyear warranty

414D  419D  424D  Distance measurement  Typical measurement tolerances (1)  ± 2.0 mm (3)  ± 1.0 mm (3)  Maximum measuring tolerance (2)  ± 3.0 mm (3)  ± 2.0 mm (3)  Leica GZM26 Position of the target board  50 m (165 ft)  80 m (260 ft)  100 m (330 ft)  Typical Range (1)  40 m (130 ft)  80 m / 260 ft  80 m (260 ft)  Range under adverse conditions (4)  35 m (115 ft)  60 m / 195 ft  60 m (195 ft)  The smallest unit to display  1 mm / 1/16 in  6/30/60 mm (10 / 50 / 100 m)
 ∅ The distance of the laser spot  6/30/60 mm (10 / 50 / 100 m)
 6/30/60 mm (10 / 50 / 100 m)
 Tilt measurement  Laser beam measurement tolerances (5)  no  no  ± 0.2°  Tolerance of case measurement (5)  no  no  ± 0.2°  range  no  no  360°  general  Laser level  2  Laser type  635 nm, <1 mW  Degree of protection  IP40  IP54  Laser cut off automatically  After 90 seconds  Automatic shutdown  After 180 seconds  Battery life (2 x AAA) 1.5 V NEDA 24A / IEC LR03  Up to 3,000 Measurement results
 Up to 5,000 Measurement results
 Dimensions (H x W x L):  116 mm length 53 mm width 33 mm depth
 127 mm length 56mm width 33 mm depth
 127 mm length 56mm width 33 mm depth
 Weight (with battery)  113 g  153 g  158 g  Temperature range: Storage operation  25 ° C to + 70 ° C (13 ° F to +158 ° F)
0 ° C to +40 ° C (32 ° F to +104 ° F)
 25 ° C to + 70 ° C (13 ° F to +158 ° F)
10 ° C to +50 ° C (14 ° F to +122 ° F)
 Calibration cycle  Not applicable  Not applicable  Tilt and compass  Maximum height  3000 m  3000 m  3000 m  Maximum relative humidity  Lt; 0 & gt; F to 120 & lt; 0 & gt; F (7 ° C to 50 ° C), 85%
 Lt; 0 & gt; F to 120 & lt; 0 & gt; F (7 ° C to 50 ° C), 85%
 Lt; 0 & gt; F to 120 & lt; 0 & gt; F (7 ° C to 50 ° C), 85%
 Safety  IEC Standard Number 610101: 2001 EN608251: 2007 (Class II)
 Electromagnetic Compatibility (EMC)  EN 55022: 2010 EN 6100043: 2010 EN 6100048: 2010
 (1) Suitable for 100% target reflectivity (white wall), low background lighting intensity, 25 ° C. (2) for 10% to 500% of the target reflectivity, high background lighting intensity, 10 ° C to +50 ° C. (3) The tolerances apply from 0.05 m to 10 m and the confidence level is 95%. The maximum tolerance may be less than 0.1 mm / m at distances of 10 m to 30 m and tolerances of less than 0.15 mm / Lt; / RTI & gt; (4) Suitable for 100% target reflectivity with background illumination from 10'000 to 30,000 lux. (5) After user calibration, the extra angle in each quadrant causes a deviation of ± 45 °, ± 0.01 ° per degree, for room temperature, and the maximum value deviation increases by ± 0.1 ° for the entire operating temperature range.
     416D and 411D laser range finder provides: Reduce estimation errors, saving time and costs Using advanced distance measurement laser technology Instantaneous measurement of the distance between two objects with onebutton operation The target is easily aligned by a bright laser beam Measurements up to 30 m (100 ft) in 3 mm (0.12 in) accuracy Distance formulas are quickly calculated, such as area (square feet per square meter) and volume Easy distance addition and subtraction calculation Indirect measurements can be made using the Pythagorean function and two other measured values Automatic power off function to extend the battery life The display hold function makes it easier to view in hardtoreach areas Battery: two AAA A carrying pocket that can be secured to the tool holder Twoyear warranty 