- Brand: LEIERDA
- Model: YN-100
- Item: YN-100
- Color Classification: 0-0.1Mpa 0-0.16Mpa 0-0.25Mpa 0-0.4Mpa 0-0.6Mpa 0-1Mpa 0-1.6Mpa 0-2.5Mpa 0-4Mpa 0-6Mpa 0-10Mpa 0-16Mpa 0-25Mpa 0 -40Mpa 0-60Mpa 0-100Mpa -0.1-0Mpa -0.1-1.5Mpa Customization -0.1-0.3Mpa Customization -0.1-0.5Mpa Customization -0.1-2.4Mpa Customization -0.1-0.9Mpa Customization -0.1-0.15Mpa Customization
- Decoration and construction content: hydropower projects
- The instrument must be mounted vertically and at the same level as the measuring point, otherwise the additional error due to the liquid level difference shall be accounted for accordingly.
- Handling, installation should avoid collision and vibration, including environmental and contact equipment, piping and so on.
- Place, use the ambient temperature if the deviation is too large, Instrument should be in the ambient temperature-40℃~+70℃, relative humidity less than 85%of Additional temperature error must be considered.
- The instrument should not exceed the upper limit of measurement when measuring steady-state loads 2/3, Measuring fluctuation pressure shall not exceed the upper limit of measurement 1/2, The minimum pressure, in both cases, should not be lower than the upper limit of measurement 1/3The measured pressure should be maintained at about two - thirds of the full scale of the meter.
- In the measurement of corrosive media, the media may be crystallized, viscous medium and large fluctuations in fluctuation pressure should be added isolation devices and buffer devices.
- Instrument storage environment The surrounding air can not contain harmful impurities that can cause corrosion, and the relative humidity of the air can not be too large.
- Before and after the instrument should be properly kept, avoid collision and strong vibration.
- The instrument should be checked frequently (at least once every three months) and found to be faulty and repaired in time.
- Please specify the media to be measured each time, so that the use of better materials.