CQ-60G enhanced specular gloss meter is a newly developed the most compact, most portable models. The performance indicators have reached the requirements of national JJG696- 2002 (mirror gloss meter verification procedures) a working machine. this product is suitable for measurement of paints, coatings, inks, plastics, stone, paper, ceramic tile, ceramic, stone, metal plating and other products gloss.
l ultra-compact design can be placed in the pocket, more portable, it is currently the smallest and lightest gloss meter.
l can operate with one hand. The battery can be used continuously for more than 40 hours, about ten thousand data can be measured.
l special moisture-proof technology allows you ease of use in a variety of harsh environments.
l means that the measurement, with no wait time, numerically stable, drift-free, for quick measurements.
l long-life light source, without replacement.
l standard quartz plate: precise, wear-resistant, stable gloss value unchanged.
Products real shot:
Ultra-compact design can be placed in the pocket, more portable
The main parameters of the relevant standards:
■ Measuring range: 0 ~ 199.9Gs (gloss units)
■ Stability: less than ± 0.4Gs / 30Min; indication error: less than ± 1.0Gs
■ Power supply: 2 AAA batteries
■ spot size: 10 × 20mm; measuring port size: 20 × 40mm
■ Conditions of Use: Ambient temperature: 0 degrees to 40 degrees; relative humidity: no more than 85%
■ Host Size: (L) 1120mm × (width) 35 mm × (high) 70 mm;
■ Weight: 200 g; standard box weight: 45 g
A plastic outer box
Standard carton (including a black quartz crystal standard board)
INSTRUCTION MANUAL a
A cloth to wipe the mirror
5.1 Power On / Off: Press the power button to the right of the instrument, the LCD display has readings, indicating power has been connected, the instrument began to work, long press the left button is off mode 5.2 Scaling: Open standard carton, the instrument measuring port on the standard black glass plate, the instrument must be placed quasi-positioning frame. press and hold the middle touch switch, press the power button to boot. calibration data is completed within 1 second. calibration value in line with the number of black glass standard board. (this machine is stable write excellent, calibration can be performed once in January) 5.3 calibration: the instrument calibration standard carton was placed on display at this value should be the work of the board and the calibration value the difference is within 1.5Gs, instrument calibration is completed measuring 5.4: place the instrument on the sample, the display reading is the gloss value of the sample is 5.5 backlight: When turned on, tap the touch key to open That turn on the backlight, turn off the backlight and tap once