- Brand: ETA
- Model: G1020
- The new 100MHz (G1100, 1 / 10X) security The new 200MHz (G1200, 1 / 10X) Security The new model of the new 100MHz (G1100, 1 / 10X) (G1010, 1 / 10X) Security The new 20MHz (G1020, 1 / 10X) security The new type of safety of the new high-voltage probe 2500V (G3100, 100X, 100MHz)
friendly reminder:爲Avoid measurement error, be sure to test the probe before the measurement and calibration, at least in the replacement of the probe, the probe when the exchange channel, you must adjust the probe compensation.
The probe calibration method is as follows:
When connecting the probe to the square wave signal output terminal of the oscilloscope calibration, adjust the frequency compensation trimmer capacitor (on the probe BNC connector) on the probe with a non-inductive sense when the probe is not in the best condition.If the impedance does not match, The measured waveform will be deformed. * If two customers to buy a non-sense with our batch, two ground springs, four color ring * (single to buy one without accessories oh ~ o ~).