(Lamp method) sulfur content analyzer FDR-3001
The sulfur content of petroleum products tester (lamp method) is in accordance with the People's Republic of standard GB/ T380 Determination of sulfur content in petroleum products (lamp method) the requirements of the design and manufacture of the instrument applicable to press GB/ T380 standard Determination of the method specified in the Reid vapor pressure not greater than 600 mm Hg light petroleum products (gasoline, kerosene, etc.) of the sulfur content.
1, glass instrument to facilitate loading and unloading.
2, Burning lamp can be adjusted up and down as needed.
,Each gas line gas flow rate can be adjusted independently.
4 , suction the sample gas after combustion, while stirring the test solution
1 Power supply: AC220V ± 10%; 50Hz.
2, the number of samples: independent five groups, each group can be a separate test.
3, electromagnetic pump: the main electromagnetic pump independent five groups, each group vacuum pumping volume can be adjusted independently the auxiliary electromagnetic pump (mixing pump) .
4, vacuum pumping adjustment: stepless adjustment potentiometer.
5, EEP adjustment: each group the EEP independently adjusted up and down, the adjustment range of not less than 15 mm.
Structural materials: countertops the cuvette holder made of stainless steel material.
7, the test tube bracket: each cuvette holder independently adjustable, the adjustment range of not less than 20 mm.
Ambient temperature: -10 ℃~ 40 ℃.
9, relative humidity: ≤ 85%
10 Total power consumption: not more than 150W