Manual spring tension and compression testing machine
Product name: Manual spring tension and compression testing machine
1, the test machine name: digital spring tension and compression testing machine
2, the implementation of standards: JBT 7796-2005 (spring tension and compression testing machine), JBT 6147-2007 (Tester packaging, packaging and marking, storage and transportation technology requirements), GBT 2611-2007 (General requirements for testing machines)
3, Model: TLS-S10-5000N
4, the test machine structure:
3.1 Host: single-column structure
3.2 Work: Handle afterburner to achieve tension and compression
3.3 Operation and display:
① lifting movement adjustable seat space (manual), axis movement to achieve the test rack stroke (manual)
② LCD character display with LCD display test force and displacement, the key input data. Micro-printer print test results.
5, the main purposes: spring tension and compression testing
6, the main technical indicators and functions:
5.1 Maximum test force: 5000N minimum readings 0.01N
5.2 Maximum tensile test space (the distance between the two hooks): 220㎜
5.3 maximum compression test space (the distance between the two pressure plate): 200 mm
5.4 Maximum test stroke: 60 mm minimum displacement readings: 0.01mm
5.5 pressure plate size: Φ90mm
5.6 test force effective measuring range: 2-200N, Range: 0-200N
5.7 Test Force showed the error: ≤ ± 1%
5.8 Displacement error (m): ≤ ± (50 + 0.15λ)
5.9 Power supply voltage: AC220V ± 10% 50HZ
5.10 Dimensions (L * W * H): 470 * 300 * 740㎜
5.11 Test Weight: about 50 kg
6. Test machine working conditions:
6.1 Ambient temperature 00C ~ 350C, relative humidity ≤80%;
Around 6.2 non-corrosive media, no strong magnetic field, no vibration;
6.3 supply voltage fluctuations should not exceed 10% of the rated voltage, a reliable ground.
7, the testing machine main configurations:
7.1 Host homemade
7.2 programmable amplification system set
7.3 Domestic high-precision displacement sensor
7.4 Domestic precision force sensor
8, quality assurance and service
1, the product complies with the national measurement standards.
2, the product guarantee period of one year, to enjoy life-long service.
3. Product delivery: Spot
4, unfinished details, negotiated settlement.
5, Our products Three Guarantees, one year guarantee period, and lifetime maintenance; warranty period of non-human cause hardware damage, I plant a free replacement, non-warranty I plant only the fees; the user's inquiry 2- 4 hours replies, we need to send sales staff to 24-72 hours to reach the scene.