The company operating the FT-105 Apparent density analyzer, heap densitometer, plastic, quality assurance, welcome to consult the discussion.
FT-105 Apparent density analyzer
FT-105 Apparent density analyzer The instrument is mainly used to set aside provisions funnel Determination of apparent density of plastic bulk material, measured apparent density for PVC resin
FT-105 Apparent density analyzer The chemical industry as well as plastic profile factory quality inspection of raw materials into the plant ideal testing equipment. Products comply with GB / T3402, GB / T1636 and other standards.
FT-105 Apparent density analyzer Technical Parameters:
The main components: base, cage, funnel, flapper, by the barrel
By the cylinder volume: 100ml ± 0.5ml
Sample volume: 110ml-120ml
Funnel angle: 40 degrees, a pore diameter of 8mm
To meet K-6721 (1966) of the requirements
The instrument is mainly used for determination of apparent density plastic outflow of bulk material from a predetermined funnel, is the chemical industry as well as plastic profile factory quality testing of raw materials into the plant ideal testing equipment. Products comply with GB / T3402, GB / T1636 GB / T20022 -2005 GB / T13173-2008 and 1636-79 standards.
Apparent density analyzer
This method is suitable for measuring the apparent density of the powder molding compound, the mass per unit volume for molding plastic packaging, mold cavity and extrusion screw design provides parameters in line with GB / T 3402, GB 1636 standards.
Apparent density analyzer
1, Sample: extracting a sample standard product requirements.
2, the test instrument:
(1) balance: a sense of the amount of 0.1 g
(2) surface view tester
(3) The cylinder-type cup: 100 ml
Note: Specific parameters, please keep in touch with us, as well as related products.
Configuration: ruler; brush, base, cage, funnel, flapper, by the barrel, manual, certificate, warranty card and so on.
Ricoeur Instruments powder powder test can also provide relevant solutions
1, the operation method
a, open the box, remove the meter, according to the packing list inventory components.
b, the detector is placed on a stable work surface.
c, with flapper Chazhu stainless steel lower funnel outlet and put away by the funnel below feeders, coaxial with the funnel.
d, with a graduated cylinder to take samples 110ML-120ML, slowly into the funnel.
e, quickly removing card, allow the sample to leave out all the natural (by feeder filled with sample shall be subjected to vibration).
f, on the feeder with a ruler scrape off excess sample by mouth, then placed on the drug weighing scales, accurate to 0.2g (measured sample may not duplicate assay).