| Weight: 1 (Kg) || Type: Tester |
| Usage: FL4-1 liquidity and bulk density tester, bulk density measurement || Custom processing: |
| Dimensions: 1 (mm) || weight: |
| Types of: || Product Usage: |
| Custom processing: || Dimensions: |
FL4-1 flowability and bulk density measuring instrument (brass)
FL4-1 flowability and bulk density measuring device (Brass) The instrument instead of GB1479-84, GB1482-84 Standard Specification for production according to GB 1482-2010-T metal powder flowability measurement standard funnel (Hall flow meter). The detection methods and results and international common The detection method has versatility. This product is also applicable to similar powder powder flowability and bulk density measurement
5.1 Emery standard sample selection, particle size less than about 106 μ m When the method in accordance with the provisions of this standard be tested on a standard funnel liquidity of the standard sample should be 4 0 ± 0.5s / 50g.
5.2 Funnel been calibrated at the factory. When you start using the current user must be calibrated with a standard sample, measured correction factor standard samples should advance in air 10 At a temperature of 5 ± 10 ℃ drying 30min, then placed in a desiccator to cool to room temperature then measuring the flow of the sample Buju measuring standard sample, six times repeated measurements, in addition to the first, the other five measurements taken Arithmetic average. In general, the requirements for the maximum difference between the results of each measurement shall not exceed 0.4s. Ends assumed 50g standard sample from a standard funnel flow time is 40s, the correction coefficient of the funnel is 40.0 divided by 5 times the measurement results arithmetic mean.
5.3 Funnel in the course of a standard sample should be regularly proofread flow time if a standard sample in the funnel vary with time should be measured to calculate the average new new correction factor.
Note: 1, In the new proposal is adopted before the correction coefficient, the reason to study change. If the time is shortened, described by repeated use of the funnel, holes due to wear powder becomes large, and prove new correction number is correct. On the contrary, Director of the flow of time, indicating a soft powder attached to the wall of the hole should be carefully remove these deposits, then repeat the test.
2, When the funnel period of time, with the 50g standard sample measured flow time reduced to 37s or less, the funnel should be scrapped and replaced with a new funnel.
Albert Instrument Co., Ltd. Ningbo Jiangbei Ricoeur
(Referred to as 'Ricoeur instrument' English: Ningbo Rlco instrument Co., LTD)
RICO Ricoeur 'began in 1999, the Department of Ricoeur instrument enterprise systems; Ricoeur instrument enterprise system is mainly engaged in analytical instruments, precision testing equipment and advanced detection equipment research and development, production, sales and after-sales service for the integration of the Industrial Company.
Which owns 'RICO Ricoeur' trademark, the products meet the requirements of international standards of GB, CCC, USP, ISO, BP, EP Pharmacopoeia.
Companies' quality-oriented 'service first' operating principles. Meet the needs of our customers. Companies in the 'science and technology, creativity, innovation, progressive' entrepreneurial spirit, the introduction of advanced production equipment and manufacturing process, through continuous technology to improve and perfect the management, strive to provide customers the best quality products and service!
Our products are widely used in chemical, petroleum, pharmaceutical, pharmaceuticals, paints, batteries, ceramics, non-metallic powder material, metal oxide powder, molecular materials, metallurgy, plastics, construction materials, soil, dyes, abrasives, food additives, catalysts, etc. related industries, provide the most complete product testing solutions and technical information for the production enterprises, research institutes and universities.
FL4-1 flowability and bulk density measuring device For measuring powder flowability and bulk density.
Under funnel diameter: φ2.5㎜ and φ5.0㎜
Angle: 60 degrees
Capacity: 25 ml
basic configuration: The stand: 1 set; funnel: (Φ2.5mm, Φ5.0mm) each one; cup; (Φ31mm, 25ml) 1 只; stopwatch 1; overflow trays 1; brush 1; ruler 15cm1 the manual ; warranty card; certificate.
Ricoeur instrument tips, as well as associated products FT-102A vibration funnel bulk density tester, FT-100A tap density tester; FT-101 Scott capacity meter bulk density tester, FT-102 Hall flow meter / Bulk Density Tester - stainless steel ; FL4-1 Hall flow meter / powder flowability tester - brass
Ricoeur Albert Instrument Also produces the following related products: Powder flowability tester / Bulk Density Tester / Hall flow meter / tap density tester / Scott capacity meter bulk density tester / bulk density Tester / aluminum powder Bulk Density Tester / measuring angle of repose instrument / angle of repose tester / apparent density analyzer / / aluminum fluoride (Alumina) Mobility analyzer, etc.
Heap densitometer, bulk density meter, a tap density gauge, Scott capacity meter, Hall flow meter, bulk density meter, aluminum powder bulk density analyzer, repose angle measurement, angle of repose, and the tap density determination instrument, the powder flowability tester, apparent density analyzer, aluminum fluoride (alumina) flow tester, bulk density meter, voltage drop tester, low DC resistance tester