The company operating the vibratory bulk density meter, magnetic oxide, quality assurance, welcome to consult the discussion.
FT-102A Zhen Fixed funnel Bulk Density Tester : According to the standard : Determination of bulk density of SJT 10215-1991 magnetic oxide powder density measuring - Vibration funnel method and measured according to ISO 3923/3 using metal powder bulk density Part III: Vibration funnel method, to meet GB / T5061-1998 metal powder bulk density determination Part 3 Vibration funnel method;
FT-102A Zhen Fixed funnel Bulk Density Tester: Scope This standard applies not freely flow through the aperture for the determination of magnetic oxide powder bulk density of 5mm. Does not apply easily broken during vibration powder, such as needles and fibrous, while applicable to similar powder bulk density measurements.
FT-102A Vibration funnel Bulk Density Tester : Instrument Principle The powder was loaded into a hopper vibration device, under specified conditions of vibration, so pink
End from the hopper by a certain height free fall, filled with a known volume of a cylindrical cup. In the bulk state, the mass per unit volume of the powder in order to represent
Bulk density powder
FT-102A Zhen Fixed funnel Bulk Density Tester: Technical Parameters
1, Funnel aperture of 7.5 mm, on the surface from the bottom of the hole and the cylindrical funnel cup is 25mm.
2, the vibrator power frequency 50Hz, funnel horizontally vibrations.
3, cylindrical cylinder capacity of 25 persons 05cm. Funnel taper 60 °
4, the funnel and cylindrical cup made of brass, and should have a certain wall thickness and hardness
5, Yao Ping should have enough weighing range, and precision weighing with disabilities can achieve 0.05 g. (User-supplied)
FT-102A Vibration funnel Bulk Density Tester
1. Determination of bulk density of powder density measuring - Vibration funnel method and reference were determined by ISO 3923/3 metal powder bulk density Part III: Determination of vibration funnel method to meet GB / T5061-1998 metal powder bulk density Part 3 Vibration funnel method;
2, the product to meet the easy maintenance, safe, reliable, economical and practical aesthetic requirements.
3, equipment installation, commissioning and use
4, confirm the ready placement of the instrument bench is level.
Installation equipment .
1, the damping pads placed under the instrument
225ML container plus a base reference to funnel spout placed in the center.
4, adjusting to the minimum vibration controller
5, turn on the power to start
6. The test can be completed in accordance with the standard requirements
Ricoeur Instruments Tips, as well as associated products FT-102A vibration funnel bulk density tester, FT-100A tap density tester; FT-101 Scott capacity meter bulk density tester, FT-102 Hall flow meter / Bulk Density Tester - stainless steel ; FL4-1 Hall flow meter / powder flowability tester - brass