The company Microcomputer Powder Density Tester, tap density meal, quality assurance, welcome to consult the discussion.
FZS4-4 Economical tap density tester
FZS4-4 Economical tap density tester (Original FZS4-4B)
FZS4-4 Economical tap density tester Production requirements of the instrumentation system according to international standard ISO3953 requirements of production, GB T 5162-2006 (determination of metal powder tap density) / satisfied, and with reference to the United States Pharmacopoeia independent research and manufacture of the powder density test instruments for measuring vibration powder real density consistent with USP (United States Pharmacopoeia), BP (British Pharmacopoeia) and EP (European Pharmacopoeia) Specification requirements, the powder through the tap, its density increases the percentage is a comprehensive reflection of the powder variety of performance. The instrument is by using a rotating eccentric slide up and down to make directional slider, percussion drill seat, the inner cylinder powder gradually tap means, mainly used for the determination of various shapes powder tap density, non-metallic powder can also be learn to use.
FZS4-4 Economical tap density tester Technical Parameters
Vibration stroke tap apparatus 3 ± 0.1mm
Vibration frequency 250 ± 15 beats per minute
Glass cylinder volume is 100 ± 1.0ml, graduated some high 175 ± 10mm. Glass cylinder volume of 25 ± 0.5ml scale section of a high 135 ± 7mm. 100ml graduated cylinder volume between the smallest scale is 1ml, measurement accuracy of ± 0.5cm ; cylinder between 25cm minimum scale volume of 0.2cm, measurement accuracy of ± 0.1cm
There Related Products: FT-100B Practical Tap densitometer ; FZS4-4 Economical tap density tester; FT-100A Microcomputer Powder Density Tester , FT-104B Angle of repose tester
FT-100A Microcomputer Powder Density Tester, USP tap density meter
FT-100A Powder Density Tester is in line, BP (British Pharmacopoeia) and EP (European Pharmacopoeia) Specification requirements and GB / T 5162-2006 / ISO3953 USP (United States Pharmacopoeia): 1993 (Determination of metal powder tap density) The instrument covers the national standard GB / T 5162-2006 / ISO3953: 1993 in various indicators.
FT-100A Powder Density Tester by a variable speed motor, vibration components, micro and mini printers and other components. And the instrument with a compact, solid, simple operation, suitable for non-metallic powder, metal powder, pharmaceuticals, chemical, batteries, abrasives as well as research and teaching, is the ideal instrument powder industry tap density test. The basic working process tap density tester is the powder cylinder fixed to the mechanical vibration device, vibration motor driven mechanical vibration device up and down vibration, graduated tube with powder containing mechanical vibration device rhythmic vibration and, with increasing frequency of vibration, after measuring cylinder powder gradually tap, vibration frequency reaches the set number of times, mechanical vibration device to stop vibration, read the volume scale test tube, according to the definition of density: mass divided by volume, thereby obtaining powder tap density after.
FT-100A Powder Density Tester Technical Specifications:
1. Sample weight: ≤500.00 g;
2. The sample volume: ≤250.00 milliliter;
3. Single frequency vibration: 0 ~ 99999 times (when set to 0 when the result is output as 'bulk density');
4. Allow the motor torque: 0.86 N.m;
5. Vibration frequency: 0 to 300 rev / min (adjustable);
6. The vibration amplitude: maximum 14mm, 1-14mm adjustable integer;
7. Repeatability error: ≤1%;
8. accuracy error: ≤1%;
9. Input Power: AC 220V (50Hz / 60Hz), 40W.