The company operating the powder flowability tester bulk density tester, quality assurance, welcome to consult the discussion.
FT-103 Natural Accumulation bulk density meter , Dust accumulation densitometer
FT-103 Natural bulk density meter stacking performance standards
It is based on national standards GB / T 16913.3-1997 provisions of design and production. The device is suitable for the determination of bulk density of dust.
Ricoeur instrument company also produces the following series : Powder flowability tester / Bulk Density Tester / Hall flow meter / tap density tester / Scott capacity meter bulk density tester / bulk density meter / aluminum powder Bulk Density Tester / angle of repose Tester / angle of repose tester / apparent density analyzer / tap density tester / aluminum fluoride (Alumina) Mobility analyzer
FT-103 Natural Accumulation bulk density meter Please refer to the sample collection GB / T 16913.1-1997 dust Physics Experiment Part 1: Dust samples collected in the relevant regulations.
FT-103 Natural Accumulation bulk density meter Principle: Dust from the funnel mouth full of free fall at a certain height measuring cup; mass per unit volume of dust inside the cup measured bulk state, that dust bulk density.
FT-103 Natural Accumulation bulk density meter Structure: funnel (flow outlet diameter 12.7mm)
Funnel made of stainless steel, which funnel taper of 60 ° ± 0.5
Ricoeur instrument prompts you: You may also use the following package of products : FT-103B dust repose angle measurement; FT-103 - Natural bulk bulk density meter; FT-104B repose angle measurement
Dust sampling experiments should be consistent GB / T 16913.1-1997 The provisions of Registration dust sampling conditions.
3.2 Dust samples in 105 ℃ dried 4 Hours after placing indoor natural cooling through 80 Mesh sieve to debris, ready determination.
Note: For less 105 Chemical reaction occurs when ℃ or melting, sublimation of dust, the drying temperature should be reduced accordingly.
3.3 The measuring device to adjust the level.
3.4 Stick stuffed with jam funnel spout. Measure 120ml The sample is poured into the funnel.
3.5 Pull out the stopper so that dust naturally fall to the lower cylinder in until after all the dust out of the funnel with a scraper to scrape the dust accumulated in the upper part of the cylinder.
3.6 The cylinder is equipped with dust sample was weighed on the balance weight.
3.7 Dust above procedure was repeated three times the bulk density is calculated according to the formula:
Where: ρ0 - Dust bulk density. g / cm3 ;
m1 , m2 , m3 - Measuring the quality of secondary dust were weighed. g ;
V - Cylinder volume. 100ml ;
continuous 3 Quality dust determinations resulting difference between maximum and minimum values should be less than 1g Otherwise, for the first 4 (Article 5 First 6 ) Determinations, until the difference between the maximum value and the minimum value is less than 1g , Take to meet the requirements of 3 Measurements average as measurement results.
FT-106 Ordinary abrasive The bulk density meter, Abrasive Bulk density meter
FT-106 Ordinary abrasive The bulk density tester (original DMP-II )
FT-106 Ordinary abrasive The bulk density tester According to international standard ISO9136: 1989 (Determination of bulk density coarse grains) and industry standard JB / T7984.2-1999 (Ordinary coarse grit abrasive Determination of bulk density) of the relevant technical requirements of development and production of special detection equipment, while meeting the GB / T20316.1-2009 bulk density tester requirements; suitable for detecting manufacturing bonded abrasives and coated abrasives bulk density of coarse particles of abrasive materials, but also for the detection of other granular material bulk density of the detection method. results and international common detection method has complete versatility.
FT-106 Ordinary abrasive The bulk density tester working principle
The abrasive test load funnel, open the drain valve, abrasive natural port along the funnel and into the measuring cylinder fall, and will fall into the measuring cylinder abrasive Calibrating the balance Weigh Calibrating the measuring cylinder abrasive quality, and then divided by the volume measuring cylinder, can be measured bulk density of abrasive measured.
FT-106 Ordinary abrasive The bulk density tester Technical Parameters
The main technical parameters
1.1 Funnel Material: Stainless Steel;
1.2 funnel Dimensions: Height 240mm; top diameter 160mm; outlet diameter: (20 ± 0.5) mm;
Catchy from the bottom of the funnel and measuring cylinder: (76 ± 1) mm measuring tube size: diameter 64mm; net height 62mm measuring cylinder volume: (200 ± 0.5) cm3 measurement error: ≤ ± 0.02g / cm3.
2.1 Measurement error: ≤ ± 0.02g / cm3.
2.2 measuring cylinder volume: 200 ± 0.5cm3.
2.3 Dimensions: 230mm (length) * 250mm (width) * 425mm (H)
FT-106 Ordinary abrasive The bulk density tester
Note: Specific parameters, please keep in touch with us, as well as related products.
Ricoeur instrument tips: You may be interested in the following products:
Common powder powder tester series
FT-100A Tap density tester; FT-100B tap densitometer; FT-101 Scott capacity meter Bulk Density Tester
FT-102 Hall flow meter / Bulk Density Tester - stainless steel; FL4-1 Hall flow meter / powder flowability tester - brass FT-103 bulk density tester; FT-103A activated clay bulk density determination equipment;
FT-103B Dust analyzer angle of repose; FT-104 Alumina bulk density tester;
FT-104A Repose angle measurement; FT-104B repose angle measurement; FT-104C aluminum fluoride mobility analyzer; apparent active agents tester FT-105A FT-105 Apparent density density measuring device; FT-105B slurry Table View density measuring device; FT-105CASTM plastic apparent densitometer; FT-106 abrasive bulk density tester (Original DMP-II)
FT-106A powder bulk density tester; FT-106B superhard material bulk density tester
FZS4-4 tap density tester; FT-102A vibration funnel bulk density tester
FZ4-3 Bulk density of the funnel-type vibration analyzer; FS4-2 Skoda Bengtsson bulk density tester
DMP-II ordinary Abrasive bulk density tester; FT-107 Fertilizer bulk density (Loose) Tester