FZS4-4 Economical tap density tester Production requirements of the instrumentation system according to international standard ISO3953 requirements of production, GB T 5162-2006 (determination of metal powder tap density) / satisfied, and with reference to the United States Pharmacopoeia independent research and manufacture of the powder density test instruments for measuring vibration powder real density consistent with USP (United States Pharmacopoeia), BP (British Pharmacopoeia) and EP (European Pharmacopoeia) Specification requirements, the powder through the tap, its density increases the percentage is a comprehensive reflection of the powder variety of performance. The instrument is by using a rotating eccentric slide up and down to make directional slider, percussion drill seat, the inner cylinder powder gradually tap means, mainly used for the determination of various shapes powder tap density, non-metallic powder can also be learn to use.
FZS4-4 Economical tap density tester Technical Parameters
Vibration stroke tap apparatus 3 ± 0.1mm
Vibration frequency 250 ± 15 beats per minute
Glass cylinder volume is 100 ± 1.0ml, graduated some high 175 ± 10mm. Glass cylinder volume of 25 ± 0.5ml scale section of a high 135 ± 7mm. 100ml graduated cylinder volume between the smallest scale is 1ml, measurement accuracy of ± 0.5cm ; cylinder between 25cm minimum scale volume of 0.2cm, measurement accuracy of ± 0.1cm
FZS4-4 Economical tap density tester maintenance
1 , Supply voltage should be between 200V ~ 240V, when the power supply voltage instability, or when there are glitches, should be equipped with precision purifying AC power supply.
2 , This equipment should be placed smooth, firm and avoid vibration, percussion, to prevent slipping.
3 , In the handling or movement, attention should be gently, and can not hit the instrument.
4 , When cleaning the surface of the instrument can be used wet towel dipped in detergent wring wipe the liquid can not flow into or spill into the interior of the instrument.
5 , Do not turn off power to the system instantly after opening each time opening and closing time interval should be greater than five seconds.
6 , Always check protective earth cable, etc., to ensure that all parts of the system are in good ground state.
7 , Vibration components belong to the moving parts of the linear bearing spilled into the powder, please clean with a brush regularly adding an appropriate amount to the linear bearing lubricants. There Related Products: FT-100B Practical Tap densitometer ; FZS4-4 Economical tap density tester; FT-100A Microcomputer Powder Density Tester , FT-104B Angle of repose tester
FZS4-4 Economical tap density tester Test object
Various metal powders: such as aluminum powder, zirconium powder, nickel powder, tungsten powder, tin powder, zinc powder, molybdenum powder, magnesium powder, copper powder and other precious metals, an alloy powder or metal oxide powder and the like.
Non-metallic powder: such as talc, kaolin, calcium carbonate, coal, phosphor, brucite, calcite, wollastonite, tourmaline, diamond, barite, fluorite, zeolite, boron carbide, clay, graphite, quartz, gypsum, bentonite, diatomaceous earth, zirconium silicate, alumina, mica, clay, and the like.
Other powders: such as soil, dyes, pharmaceuticals, pesticides, abrasives, paints, food additives, catalysts, cement, sand and other rivers There Related Products: FT-100B Practical Tap densitometer ; FZS4-4 Economical tap density tester; FT-100A Microcomputer Powder Density Tester
FZS4-4 Economical tap density analyzer instruction video link:
http: //v.youku.com/ v_show / id_XMzc3NDUxOTU2 html.
http: //www.tudou.com/ programs / view / UR6wdJ0OKAM /