| Uses: Determination of bulk density|| Repeatability error: 1|
| Samples can be measured state: magnesium bulk density meter, magnesium bulk density meter, density meter|| Custom processing:|
| Measuring range: bulk density tester magnesium, magnesium bulk density meter, density meter|| Weight: 12|
| Type: Density Tester|| Specifications: FT-103A activated clay bulk density|
| Scope: magnesium bulk density meter, magnesium bulk density meter, density meter|| Dimensions: 11 (mm) |
Magnesia The bulk density tester
FT-103A activated clay bulk density tester GBT 23771 bulk density meter, bulk density inorganic chemical products tester
GB 25571 bulk density meter, HGT2764 bulk density tester / sodium bulk density measuring device
FT-103A activated clay bulk density measuring device Is based on national standards GB / T 25571-2010 national food safety standards, food additive provisions of activated clay design, production. The device is suitable for the determination of food additives, activated clay bulk density, while also meet HGT2764-2008 peroxide sodium carbonate industry industry standards for bulk density measuring device requirements and conform GB210.2-2004. Also suitable for the determination of inorganic chemical products GBT 23771-2009 bulk density. HG / T2573-2006 industrial light magnesia (HG T2573-2006 instead of HG T2573- 1994) Determination of the bulk density requirements
FT-103A activated clay bulk density measuring device Principle: Dust from the funnel mouth full of free fall at a certain height measuring cup; mass per unit volume within the next measuring cup bulk state, that is the bulk density of the sample to be tested is loaded funnel natural port along the funnel and into the fall. the measuring cylinder, will fall to the sample Calibrating measuring cylinder in the balance that measure the quality of the sample tube after taking evenness, and then divided by the volume of the measuring cylinder, we can measure the bulk density of the products being tested.
FT-103A activated clay bulk density measuring device Structure: 2.1: funnel flow outlet diameter 30mm; funnel height: 135mm, funnel made of stainless steel, 2.2: cylinder diameter: 80mm, volume: 500ml
Ricoeur Instruments Inc. The company's products have the following eight series: Precision wire (Harness) Tester, harness (wire) conduction tester series; wire and cable tester with a full set of test equipment series; a full-power wire tester and test equipment series;; mechanical equipment and test equipment series; safety equipment series; retardant material testing equipment series; direct reading of solid / liquid density meter series; powder powder testing instrument series.
1. Preparation of sample: The sample to be tested is placed in an oven at 1100 C and dried at a temperature of 1 hour, cooled to room temperature before measurement.
2. Use a 250mL beaker filled with test sample. Close the hopper discharge port and export center measuring cylinder is placed. The sample is poured into the funnel, open the discharge valve of the funnel abrasive free fall until all the stream After this process should avoid vibration.
3. has been measuring ruler scrape excess specimen tube, when scraper ruler and measuring about 45 barrels catchy plane0 Angle, perpendicular scratch four times. This process should prevent compaction measuring cylinder in the sample, to avoid vibrations. Then measure the quality of the bucket weighing samples.
4. Bulk density is calculated as follows:
Where: P ap - Bulk density of the sample to be tested, unit: g / cm3;
m1 - Quality measurement sample cylinder, unit: g;
V- measuring cylinder volume, unit: cm3.
5. A sample measurement should be repeated three times, the arithmetic average of the three measurements as a result of the test sample bulk density values.
6. The results of the operation were measured using the same instrument repeat the same, the error should not exceed ± 0.02g / cm3.
FT-103A activated clay bulk density tester Operation video link:
http: //www.tudou.com/ programs / view / f8Xt5vkBzyw /
FT-103 bulk density meter natural accumulation. Dust accumulation densitometer
FT-103 Natural bulk density meter stacking performance standards
It is based on national standards GB / T 16913.3-1997 provisions of design and production. The device is suitable for the determination of bulk density of dust.
Ricoeur instrument company also produces the following series : Powder flowability tester / Bulk Density Tester / Hall flow meter / tap density tester / Scott capacity meter bulk density tester / bulk density meter / aluminum powder Bulk Density Tester / angle of repose Tester / angle of repose tester / apparent density analyzer / tap density tester / aluminum fluoride (Alumina) Mobility analyzer
FT-103 bulk density meter natural accumulation Please refer to the sample collection GB / T 16913.1-1997 dust Physics Experiment Part 1: Dust samples collected in the relevant regulations.
FT-103 bulk density meter natural accumulation Principle: Dust from the funnel mouth full of free fall at a certain height measuring cup; mass per unit volume of dust inside the cup measured bulk state, that dust bulk density.
FT-103 bulk density meter natural accumulation Structure: funnel (flow outlet diameter 12.7mm)
Funnel made of stainless steel, which funnel taper of 60 ° ± 0.5
Ricoeur instrument prompts you: You may also use the following package of products : FT-103B dust repose angle measurement; FT-103 bulk density meter repose; FT-104B repose angle measurement
Dust sampling experiments should be consistent GB / T 16913.1-1997 The provisions of Registration dust sampling conditions.
3.2 Dust samples in 105 ℃ dried 4 Hours after placing indoor natural cooling through 80 Mesh sieve to debris, ready determination.
Note: For less 105 Chemical reaction occurs when ℃ or melting, sublimation of dust, the drying temperature should be reduced accordingly.
3.3 The measuring device to adjust the level.
3.4 Stick stuffed with jam funnel spout. Measure 120ml The sample is poured into the funnel.
3.5 Pull out the stopper so that dust naturally fall to the lower cylinder in until after all the dust out of the funnel with a scraper to scrape the dust accumulated in the upper part of the cylinder.
3.6 The cylinder is equipped with dust sample was weighed on the balance weight.
3.7 Dust above procedure was repeated three times the bulk density is calculated according to the formula:
Where: ρ0 - Dust bulk density. g / cm3 ;
m1 , m2 , m3 - Measuring the quality of secondary dust were weighed. g ;
V - Cylinder volume. 100ml ;
continuous 3 Quality dust determinations resulting difference between maximum and minimum values should be less than 1g Otherwise, for the first 4 (Article 5 First 6 ) Determinations, until the difference between the maximum value and the minimum value is less than 1g , Take to meet the requirements of 3 Measurements average as measurement results.
4, the test results
Please fill in the test report: The test report shall include the following:
a. Necessary to explain the identification of the sample;
b. Date, measuring personnel, measurement methods, measurement report;
c. Report indoor weather conditions and determination of bulk density of dust;
FT-103 bulk density meter natural accumulation operation of the video connection: http: //www.tudou.com/ programs / view / jkVHJC0CQKM /
FT-106 ordinary abrasive bulk density tester, abrasive bulk density meter
FT-106 ordinary abrasive bulk density tester (Original DMP-II)
FT-106 ordinary abrasive bulk density tester According to international standard ISO9136: 1989 (Determination of bulk density coarse grains) and industry standard JB / T7984.2-1999 (Ordinary coarse grit abrasive Determination of bulk density) of the relevant technical requirements of development and production of special detection equipment, while meeting the GB / T20316.1-2009 bulk density tester requirements; suitable for detecting manufacturing bonded abrasives and coated abrasives coarse particles ordinary abrasive bulk density, but also for the detection of other granular materials Bulk density Its detection methods and results and generic detection methods internationally with full versatility.
FT-106 ordinary abrasive bulk density tester working principle
The abrasive test load funnel, open the drain valve, abrasive natural port along the funnel and into the measuring cylinder fall, and will fall into the measuring cylinder abrasive Calibrating the balance Weigh Calibrating the measuring cylinder abrasive quality, and then divided by the volume measuring cylinder, can be measured bulk density of abrasive measured.
FT-106 ordinary abrasive bulk density tester Technical Parameters
The main technical parameters
1.1 Funnel Material: Stainless Steel;
1.2 funnel Dimensions: Height 240mm; top diameter 160mm; outlet diameter: (20 ± 0.5) mm;
Catchy from the bottom of the funnel and measuring cylinder: (76 ± 1) mm measuring tube size: diameter 64mm; net height 62mm measuring cylinder volume: (200 ± 0.5) cm3 measurement error: ≤ ± 0.02g / cm3.
2.1 Measurement error: ≤ ± 0.02g / cm3.
2.2 measuring cylinder volume: 200 ± 0.5cm3.
2.3 Dimensions: 230mm (length) * 250mm (width) * 425mm (H)
FT-106 ordinary abrasive bulk density tester
Note: Specific parameters, please keep in touch with us, as well as related products.
3.1. The instrument should be installed in a stable work surface.
3.2. The adjustment of three levels of adjustment screw on the chassis so that the chassis is level.
3.3. Adjust the height adjustment screw, so that the distance from the end face of the lower end surface of the measuring tube funnel maintained in a predetermined range (76 ± 1mm) within.
3.4. Adjust the height adjustment screw, change the direction of the funnel, so that the center line of the funnel and measuring cylinder coincide.
3.5. After the installation is complete, the instrument should avoid strenuous vibration.
FT-106 ordinary abrasive bulk density tester operation of the video connection:
http: //www.tudou.com/ programs / view / rIp5b3SclZs /
Ricoeur instrument tips: You may be interested in the following products:
Common powder powder tester series
FT-100 tap density tester; FT-101 Scott capacity meter Bulk Density Tester
FT-102 Hall flow meter / Bulk Density Tester - stainless steel; FL4-1 Hall flow meter / powder flowability tester - brass FT-103 bulk density tester; FT-103A activated clay bulk density determination equipment;
FT-103B dust repose angle measurement; FT-104 alumina bulk density tester;
FT-104A repose angle measurement; FT-104B repose angle measurement; FT-104C aluminum fluoride mobility analyzer; apparent active agents tester FT-105A FT-105 Apparent density density measuring device; FT-105B slurry was apparent density measuring device; FT-105C ASTM plastic apparent densitometer; FT-106 abrasive bulk density tester (Original DMP-II)
FT-106A powder bulk density tester; FT-106B superhard material bulk density tester
FZS4-4 tap density tester; FT-102A vibration funnel bulk density tester
FZ4-3 bulk density of the funnel-type vibration analyzer; FS4-2 Skoda Bengtsson bulk density tester
DMP-II ordinary abrasive bulk density tester; FT-107 fertilizer bulk density (Loose) Tester