| I. Features: |
ZRN506 series torque sensor is designed to transmit torque of the torque sensor in the standard parameters measured on the basis of torsional strain sensor developed.
ZRN506 type torque and rotational speed sensor is ZRN505 series of dual flange form a torque sensor, without base mounting. Easy installation, compact size.
This series torque sensor includes:
1. Torque measurement functions;
2. Energy and the non-contact signal transfer function.
3. The output signal of the digital functions.
4. You can pass static torque signal, rotation torque signal, dynamic torque signal, a static torque signal.
And whether the rotation speed and direction unrelated 5. transfer signal can be adapted for a long time, high-speed operation.
6. High accuracy, good stability;
5. Small size, light weight, easy to install;
7. without repeatedly reversing the zero torque can be continuously transmitted signal;
8. No collector rings and other wear parts, can be as high speed long-running;
9. Strong anti-interference; can be anywhere, any orientation;
Second, technical indicators
1. range: 510205010020050010002000N · m, 10000N m;, five non-standard range of dimensions and size can be customized.
2. Accuracy: 0.5% F · S;
3. For Speed 3000 rev / min.
4. Ambient temperature: 0 ℃ -50 ℃
5. Frequency Response: 100 μs
7. Output signal: 0-12v square wave frequency; load current (10mA zero torque:. 10 KHZ; Forward span: 15 KHZ; reverse full scale: 5 KHZ;
8. The signal receptacle: (1) 0V (2) + 15V (3) -15V (4) blank (5) torque signal....
1. ZRN506 Double flange torque sensor from the stationary housing and the rotating disk of two parts.
2. Adoption of the flange coupling are coupled with the power or load.
3. Flexible stationary housing fixed to the base. Allow swing, can not rotate.
Four, signal processing
1. Use ZRN505 series torque meter displays the torque value.
2. Direct torque frequency signal to a computer or PLD processing;
3. Convert to 4-20mA or 1-5V analog signal.
1. The wiring must be correct;
2. ± 15V range: not less than ± 14.5V shall not exceed ± 15.5V;
3. The output signal line may not be on the ground, short to power, the output current is less than 10mA;
The shielded cable shield must be connected to the common terminal ± 15V supply (power ground).
5. Use the if in doubt please contact the company, may not disassemble the warranty period.